compatible with OneLab 1.11.2 (released 31 May 2021)
To prevent false-positive detections during the tip insertion procedure with a 300uL multi-channel pipette, Andrew+ quickly checks its alignment before every tip insertion.
On Andrew+, the algorithm’s robustness to calculate the geometry of a multi-channel pipette equipped with tips has been enhanced.
“Shaker+” and “Heater-Shaker+” can both be used with profiles.
A typo in the mac address of the ethernet interface, stated in the pairing interface of every Device+, has been corrected.
The “Heater-Shaker+” - a device capable of Shaking and Heating at the same time or independently - has been integrated into the One Lab ecosystem.
Andrew+ is now compatible with 50-1200uL non-extended Sartorius tips.
Download from here 1.112.0 declaration of quality.
compatible with OneLab 1.11.0 (released 29 March 2021)
5mL and 10mL pipettes will not create automatic bottom air-cushion anymore if not required by the protocol.
The rhythm used to press the button of any Device+, Andrew+, and Pipette+ to enable its pairing mode has been unified.
When powered, the Tube-Shaker+ device raises its temperature and partially heats the consumables installed. This will not happen anymore.
A new vacuum pump (IKA Vacstar Control) can now be connected and used with Vacuum+, allowing an active control of the pressure via OneLab.
Download from here 1.110.0 declaration of quality.
compatible with OneLab 1.10.2 (released 22 February 2021)
Andrew+ speed has increased dramatically thanks to a massive redesign of its routines. The system is now, on average, 30% faster than before.
Misbehaviors in the kinematic of Andrew+ arm have been resolved in order to improve the quality of the movements and their reproducibility.
A bug that prevents the system from being able to recover itself on Devices+ in case of failure has been fixed.
Download from here 1.102.0 declaration of quality.
compatible with OneLab 1.10.1 (released 21 December 2020)
On some WiFis, the presence of some special characters in the password (like '&' or '#') was the reason behind a failure of the pairing procedure. The problem has been corrected.
Download from here 1.101.0 declaration of quality.
compatible with OneLab 1.10.0 (released 30 November 2020)
The pairing process of Andrew+, Pipette+, and Devices+ has been improved and the "pairing over ethernet" is now faster and more reliable. Some minor misbehaviors have been fixed on the "pairing over wifi" procedure.
The erratic behavior of Andrew Alliance's pipettes in handling low volumes using repetitive and reverse mode has been corrected.
Redundant actions have been removed during the execution of a protocol, resulting in an improvement of the overall execution time for some scenarios where Andrew+ and Device(s)+ interact.
A smarter evaluation algorithm has been integrated to simulate the different pipetting actions of a protocol and to override if required, some of the decisions made by the user to protect pipettes and to minimize the dead volume.
Audit logfiles reports were not compressed during the finalization of the experiment if the experiment name contained a Unicode character. This will not happen anymore.
Various other improvements and performance enhancements.
Download from here 1.100.0 declaration of quality.